Specifications include, but are not limited to: 1
Footprint must not exceed: 31”W X 76”H X 47”L
Recessed units must fit in existing wall opening.
Interior chamber size: minimum 18” X 18” X 36”, maximum 22” X 22” X 40”
Units must be all 316L stainless steel construction including the chamber and jacket.
Unit must have elliptical shaped chamber.
Unit must include single manual vertical sliding door which must be constructed of 316L stainless steel.
Unit must include two removable shelves.
Unit must use existing house steam (pressure and temperature).
Utility connections must be compatible with existing building connections without extensive reconfiguration required.
The vendor will provide on-site training. As each floor is up and running we will require a training session with each floors occupants. ~ 1 hour or as long as needed.
Units must include the following cycles: Gravity, Liquid, Pre-Vacuum, Bowie-Dick test, Vacuum Leak test.
Units must include at least 15 storable cycle programs.
Equipment must meet all UL certifications and standards or an equivalent NRTL (Nationally Recognized Testing Laboratory Program)