Specifications include, but are not limited to: The Diagnostic Ultrasound Imaging System and all attached devices shall be new and equal to
current industry imaging standards within six months, based on digital architecture, with
the most recent software available. This unit and its accompanying transducers and
attached devices shall be covered by a minimum 3 years warranty. Vendor proposal shall
include all freight charges.
The Diagnostic Ultrasound Imaging System shall be equipped with all transducers and
software for Cardiac, Vascular, Abdominal, OB/Gyn, and Superficial Structure applications
The unit must have all necessary imaging and calculations software packages for:
Vascular Technology – including Cerebral Vascular, Peripheral Vascular, Abdominal
Vascular and Transcranial Doppler, Imaging & Non-Imaging.
Echocardiography – including Adult Echo
Abdominal and Small Parts Imaging – including Abdominal, Urological / Renal, Small
Parts, Musculoskeletal and Superficial imaging and vascular Doppler
Obstetrical and Gynecological Imaging – including Obstetrical calculations packages,
OB Doppler, Gynecological Doppler
3D/4D, Multiplanar reconstruction imaging for Obstetrical and Gynecological
Strain/Speckle Imaging for Echocardiography
Panoramic Imaging capabilities for the superficial, abdominal, vascular transducers.
The base unit shall also include:
2D real time imaging, compound imaging, auto image enhancement, M-mode and Color
M-mode, pulsed Doppler, color flow Doppler, duplex and triplex mode, simultaneous Color
and CW Doppler, vascular enhancement technology, cine loop image review, integrated
tissue harmonics, ECG and respirometer with necessary cables and leads, color power
Doppler imaging or equivalent, and tissue Doppler or equivalent.
This unit shall include the following transducers:
2-4 MHz Multi-frequency Phased array or equivalent (Adult echocardiography)
2-4 MHz Multi-frequency Phased array or equivalent -Transcranial Doppler and Imaging
1-5 MHz Multi-frequency Curved linear array or equivalent (Abd/OB/Gyn/Vascular)
5-15 MHz Multi-frequency Linear array or equivalent (Vascular/Superficial Structures)
2-4 MHz Continuous wave non-imaging cardiac “ped-off” or equivalent
5-12 MHz Continuous wave non-imaging probe or equivalent – Vascular applications
1-5 MHz Multi-frequency Curved linear array automated 3D/4D Transducer (OB/Abd)
5-12 MHz Multi-frequency Highly Curved linear array automated 3D/4D Endovaginal
This unit shall also include:
Flat-Panel Multi-positional display monitor, CD-RW/DVD, USB drive capabilities for image
recording, network connectivity items, DICOM connectivity for Digital Imaging distribution
with existing archiving system, Service Manuals.